Surgical Stainless Steel Dual Functional Sliver Cannulated Drill with Twist Drill Bits (ND-2011)
|Hand-piece Unit Rise in Temperature||≤25ºC|
|Charging Time||About 3 hours|
|Radial Run-out||≤ 0.5mm|
|Charger Input Voltage||AC 220V, 50Hz|
|Charger Input Power||30VA|
|Charger Output Power||14.4V, 900mA|
2. Product Technical Merits
- Can be sterilized up to 135ºC
- Motor is imported from German
- Strong power ,low noise,small vibration
- Reduced the risk of surgery
- Variable speed and lightweight for better accuracy
- Axial run-out of concentrically is within 0.05mm.
- Multipurpose interface is easier to use in clinic
- 4.0mm hollow for all brand needles to pass through.
- Ni-MH battery without memory environment.
3. Product Display
-One drill chuck with key
-One hand piece
-One sterilizing channel
-One aluminium box
5. About Us
Wuhu Ruijin Medical Instrument & Device Co., Ltd.
Wuhu Ruijin Medical Instrument & Device Co., Ltd., established in 2007 and locates in Wuhu City, Anhui Province, China. We are a medical device manufacturer integrated with R&D, production and sales, focusing on medical drill series and micro medical electric drill series. Our main products include medical bone drill, multi-functional cannulated drill, oscillating saw, reciprocating saw, and medical electric plaster saw, which can be used in many surgeries such as Trauma Surgery, Thoracic Surgery, Joint Operation and Veterinary Orthopedic Surgery, etc.
Visiting our website: whruijin.en.made-in-china.com
No. 33 Wanchun Road, Wuhu, Auhui,China. 241000
6. Payment & Delivery
1: Price: FOB USD 450-545 / Set, Shanghai Port
2: MOQ: 1 piece
3: Payment Terms: T/T in advance or western union
4: Delivery Time: Within 3 days after payment
8. Attention & Notes
|1. The device is forbidden to be demolished by non-professionals. If any|
problem, please contact with us for after sales service.
2. Our products are chargeable, it can match for electricity source 110V60Hz-220V/50Hz.
|3. The battery needs to be taken out of the machine's mobile phone, otherwise a small automatic discharge will shorten the battery life.|
|4.The battery cannot be disinfected. The battery space should be dry after sterilization.|
|5. Saw blade, the drill bit is cleaned with a purifier and kept dry after operation|
|6. The sterilization temperature is lower than 135 degrees Celsius. If it exceeds 135 degrees Celsius, the motor will be damaged.|
|7.Professional manufacturer with 24 hours online service , warranty offered, CE marked.|
Ruijin Products General Technical Data:
|Normal bone drill RJ-0110||0-1300 r/m||≤ 65db||≤ 0.5mm||≥ 1.6NM|
|Cannulated bone drill RJ-0210||0-600 r/m||≤ 65db||≤ 0.5mm||≥ 3.5NM|
|Oscillating saw RJ-0310||0-18000 osc/m||≤ 75db|
|AcetabulumSlow drill RJ-0410||0-300 r/m||≤ 65db||≤ 0.5mm||≥ 6.5NM|
|Sternum saw RJ-1810||0-15000 ocs/m||≤ 75db|
|Craniotomy drill RJ-1510A||0-900 r/m||≤ 40db|
|Craniotomy mill RJ-1510B||0-40000 r/m||≤ 40db|
|Grinding drill (RJ-JZ-2)||0-40000 r/m||≤ 65db||≥1.25NM|